An internal, high gain, fully-differential amplifier and a 16-bit analog-to-digital converter combine to accurately digitize the input signal. The analog input is digitally filtered, to user selected sample size and filter window (band), allowing optimization for specific operating conditions, trading off depth of filtering and step change tracking. Self-calibrating +/- 10Vdc and 4-20mA analog output signals are standard. A RS232 interface provides facilities to communicate, command, query operating modes, parameters and to continuously read data.
A TEDS status indicator provides the information on the TEDS configuration of the input. See IEEE website for more
information on TEDS: https://standards.ieee.org/develop/regauth/tut/teds.pdf
A 5Vdc@ 60mA or 10Vdc@120mA, short circuit protected, transducer excitation supply is also provided.
A front panel RCAL switch provides a convenient calibration feature. TARE and CALIBRATION are easily performed via front panel pushbuttons.